Projection Microscope

The incorporation of precision glass circle, scale plate
with minutely finished rotating mechanism and tropic proof high sensitive
optical system enable the precise reading of test result without the aid
of vernier.

MEDIMEAS with the close co-operation of M/s KYOWA - JAPAN
offers a selected range of high quality Projection Microscopes suitable
for measurements and group discussions.

SP-Series Projection Microscopes incorporate the most advanced
technical features to make it versatile for a wide range of
applications in Research and Industries. The Din Standard Plan
Achromatic Optics produces uniformly images and exceptionally sharp
resolution over the entire screen.

The 195 mm round graduated screen made from the finest glass is rotate
able in 360° on ball bearing and has a fresnel
lens behind it for brilliant and uniform intensity of projection
throughout the screen area. The Co-axial mechanical stage is attached
for comfortable and convenient manipulation of slides for research
works, like checking linear measurement and comparison of contours etc.
A magnification correction knob is provided in the microscope on which
a compensating eyepiece moves to attain exact magnification in
different objectives.

The
Microscope is offered as Standard Projection Microscope Model SP-585
and SP-585A and also with Binocular Head arrangement as SP-588.

The Polavision SP-590 Microscope is offered as Standard
Polarising Microscope and also with Binocular
Head arrangments as SO-590A.

Technical Features

Optical System :

Din Standard TC Series objectives, Optionally/Plan Optical
System.

Main Body :

A most durable, sturdy & elegant designed body for utmost
convenience to use and to accept several optional accessories as a
multipurpose system microscope.

Focussing :

Coarse and fine focusing system. Coarse focus range 30mm and
fine focusing reading to 0.002 mm.

Nose Piece :

Revolving quadruple/quintuple nose piece with ball bearings
for perfect alignment

Stage:

With Co-axial Mechanical Stage low drive control, travel area
75-50mm

Condensor :

Movable Abbe Condensor
N.A. 1.25 with iris diaphrasm and filter
holder.

Illuminator :

Halogen illumination system having 12V/100W Halogen lamp with
variable lightcontrol.

Observation :

Observation on screen /optionally binocular head.

Magnification Range :

100/125x to 1000x & also upto
1500x

Specifications:

SP-585

SP-585

SP-585

SP-585

SP-585

Projection Screen

195 mm Rotatable
360°

Objectives

4x/5x,10x,

20x &

40x/45x

4x/5x,10x,

20x,40x/45x

,60x

4x/5x,10x,

20x,40x/

45x,60x

P4x/5x,P10x,

P20x&P40

x/45x

P4x/5x,P10x,

P20x&

P40x/45x

Optics

Standard

Optics

Din Standard

TC Series

Din Standard

TC Series

Din Standard Polarising
Objective

Din Standard Polarising

Objective

Eye Piece

Compensating

Compensating

Compensating

and

WF10x Paired

Compensating

Compensating

and

WF10x Paired

Magnification Range

125x to 1000x

100x to 1500x

100x to 1500x

125x to 1000x

125x to 1000x

Illuminator Halogen Lamp
12V/100W

Binocular Head

Polarising